|View printer-friendly version|
|Medtronic Announces Voluntary Recall of Diabetes Infusion Sets|
DUBLIN - September 11, 2017 - Medtronic plc (NYSE:MDT) announced today that it has started to inform patients worldwide of a voluntary recall of specific lots of infusion sets used with all models of Medtronic insulin pumps. The recall is related to a certain discontinued component in these infusion sets and does not include insulin pumps or glucose sensors.
The company determined, through recent field reports from patients and root cause analysis, that a component, the vent membrane, in the recalled infusion sets may be susceptible to being blocked by fluid during the process of priming/fill-tubing. This situation can lead to potential over-delivery of insulin shortly after an infusion set change, which may cause hypoglycemia. Currently manufactured infusion sets, available to patients since April 2017, include a design update of this component which the company believes reduces the risk of insulin over-delivery after an infusion set change. The company will work with patients to ensure recalled infusion sets with the discontinued component are returned and replaced with new infusion sets containing the updated component at no cost.
Medtronic has contacted the U.S. Food and Drug Administration (FDA), along with other regulatory bodies around the world, to share information related to this issue. Medtronic will continue working directly with government and regulatory authorities on this global voluntary recall.
"Our priority is to work with our patients to mitigate risk to patient safety. While we have shipped a significant number of the new and enhanced sets since April, we are committed to replacing recalled infusion sets for all patients," said Francine Kaufman, M.D., chief medical officer of the Diabetes Group at Medtronic. "Our Medtronic Diabetes team will work as quickly as possible to complete all exchanges to the new and enhanced set and fully support our customers throughout this process."
Customers in the United States (U.S.) can determine if they have recalled infusion sets by visiting https://checklots.medtronicdiabetes.com.
Customers outside the U.S. will receive instructions specific to their country. In Europe, Middle East and Africa (EMEA region), customers can determine if they have recalled infusion sets by visiting www.mmc.medtronic-diabetes.com/look.
Patients can always consult the advice of their healthcare professional regarding their medical treatment.
If a customer in the U.S. has experienced an issue with the use of a Medtronic infusion set, please report it to the 24-hour helpline at +1-800-204-7616. Customers can also report adverse events to the FDA's MedWatch Adverse Event Reporting program:
About the Diabetes Group at Medtronic (www.medtronicdiabetes.com)
Any forward-looking statements are subject to risks and uncertainties such as those described in Medtronic's periodic reports on file with the Securities and Exchange Commission. Actual results may differ materially from anticipated results.